J241 is only required for 10BASE-T and not the other data rates. The issue arises from the fact that for 10BASE-T, the A differential pair is used for Tx, and the B differential pair is used for Rx. This is quite different from the other data rates. The TC1 section of these test fixtures basically has a power splitter that connects the AWG signal to all four of the differential pairs simultaneously. This is convenient when testing 100BASE-T and 1000BASE-T, but has an unintended consequence when testing 10BASE-T. The AWG signal that is connected to the B differential pair disturbs the DUT, and disrupts the test. So, by applying the jumper 241, we short out the AWG signal at the output of the power splitter. The B pair then is just a non-active 100 ohm load for the DUT. This allows the return loss test to be performed on the 10BASE-T Tx lines without disruption.
This FAQ Applies to:
Product: TDS5052, TDS5054, TDS5104, TDS5034, TDS5032, TDS5054B, TDS5104B, TDS5032B, TDS5052B, TDS5054BE, TDS5034B, DPO7054, DPO7104, DPO7254, DPO70404, DPO70604, DPO70804, DSA70404, DSA70604, DSA70804, DPO7354, DPO71254, DPO71604, DPO72004, DSA71254, DSA71604, DSA72004, DPO7254C, DPO7354C, DPO7054C, DPO7104C
FAQ ID : 61071View all FAQs »