Can the P7708 with P77BRWSR measure 0-10V signal?
Yes, 10V is possible with using the P77BRWSR browser. The key specs are the differential input range and the operating voltage window.
You need to ensure that the signal being probed stays within the linear input range and this range can be moved anywhere in the operating window using the probe’s offset voltages.
Using an offset of 5V and by putting one leg of the browser (B-side) on 0V and the other (A-side) on 10V, you can keep the signal in the linear input range of the probe’s amplifier.
Here is a graphic of the input range and operating voltage window that might help
이 FAQ는 다음에 적용 됩니다:
제품 시리즈: 저전압 디퍼런셜 프로브
FAQ ID 254901모든 FAQ 보기 »