How do I adjust the LCD contrast on my TSG95?
The LCD contrast is controlled by R1 on the main board inside the instrument. Turning the potentiometer clockwise increases contrast while turning it counter-clockwise decreases contrast. Please refer to the diagram below for locating R1 on the main board:
이 FAQ는 다음에 적용 됩니다:
제품 시리즈 없음
FAQ ID 63486모든 FAQ 보기 »