Do I need to send the INIT command to trigger the Model 2430 when using Pulse Mode?
Product(s) :SourceMeter Instruments: 2430, 2430-C
Question : Do I need to send the INIT command to trigger the 2430 when using Pulse Mode?
Answer: When the Model 2430 is programmed for Pulse Mode over the bus (GPIB, RS-232), you do not need to send an :INIT command after turning the Output ON. When operating in pulse mode, sending the :OUTP ON command will also send the :INIT command to start the pulse process.
Sending an :INIT command following the :OUTP ON command will trigger the Pulse once again.
이 FAQ는 다음에 적용 됩니다:
제품 시리즈: 2400 그래픽 시리즈 SMU
FAQ ID 69616모든 FAQ 보기 »