Can I phase synchronize the signal outputs of two or more AFG3000 instruments?
Yes. Waveforms from two or more generators can be synchronized by operating one AFG310x, AFG302xB, AFG3011, or AFG325x as "Master Unit" and any of the AFG3000 Series instruments as "Slave Units." In this configuration, the master's Reference Clock and Trigger Outputs are connected to the slaves' Reference Clock and Trigger Inputs, respectively. Any signal delays introduced by the cable lengths can be compensated for with the Trigger Delay function.
See FAQ ID: 836516 for more information.
이 FAQ는 다음에 적용 됩니다:
제품 시리즈: AFG3000C 임의/함수 발생기
FAQ ID 56401모든 FAQ 보기 »