When I use a custom made arbitrary wave form, set the modulation to burst and the trigger source to external it is not possible to get any signal out of the device.
This was a known limitation on AFG1062's that had firmware v1.0.4 or lower. This limitation has been resolved with the installation of firmware v1.0.5 which can be downloaded from Tek.com.
이 FAQ는 다음에 적용 됩니다:
제품 시리즈: AFG1000임의/함수 발생기
FAQ ID 247536모든 FAQ 보기 »